AquaCalc 5000 Revision History
Advanced Firmware – H
H8c - November 2000
- Fixed bug for no velocity mesurement reporting.only 20 seconds. When taking a full 40 second measurement at a location with no velocity, AquaCalc would report only a 20 second meaasurement in the output file.
- Added user selectable sensitivity to turbulent flow, reduces resets. Useful mostly with Pygmy meters. User selectable 4:1 or 8:1 timing ratios.
H8a - Internal JBS version. Not released
H7 - Added New Standard 2 Rating Tables.
- Minor bug fixes, including a random problem with loading station zero. Added clock diagnostic feature available to repair technician only.
H5 - Stable release for several years
Basic Firmware – G
G7b - User selection of turbulent flow ratio of previous time between clicks and current time (4:1 or 8:1)
- Fixed counting problem introduced with G6. Added new counting algorithm that reduces turbulent flow resets. (2:1 timing ratio is not user selectable)
G6 - Added New Standard 2 Rating Tables. Introduced problems with counting.
G5 - Stable release for several years